FAQ: “Our site requires us to standardize the instrument every 4 hours. Is there any way to change this in the EasyMatch QC software?”
The EasyMatch QC software prompts the user to Standardize the instrument ever 8 hours, this is the default setting when the software is installed. If your company requires the instrument to be standardized at a different interval, this can be changed in the EasyMatch QC software.
To change the Standardization default setting, go to Sensor/Set Interval. A new window opens that allows you to specify the new Standardization interval in hours. Use the up and down arrow to make your changes. Click OK to save.
Mr. Philips has spent the last 30 years in product development and management, technical sales, marketing, and business development in several industries. Today, he is the global market development manager for HunterLab, focused on understanding customer needs, providing appropriate solutions and education, and helping to solve customer color challenges across these industries and cultures.